Hydrostatic liquid level sensor PS-C-01-420
Submersible liquid pressure sensor. The pressure exerted by water on the sensor is transmitted as a voltage signal (4-20 mA). The hydrostatic pressure is directly proportional to the immersion. The sensor is designed for operation in acids (hydrochloric acid, sulfuric acid, hydrofluoric acid, sodium hydroxide, hydrogen peroxide and other highly corrosive liquids, chemical and galvanic wastewater). The sensor capillary is made of F4 tetrafluoro. The signal of the measured pressure of the liquid is transmitted through a tube to the sensor, which is located in an external, airtight housing.
pic. submersible, pressure liquid level sensor PS-C-01-420.
||0 ~ 1m
|Working environment temperature
||-40 ~ +150°C
|Degree of protection
Fig. PS liquid level pressure sensor dimensions and wire markings.
Dedicated meters for sensors are MG14 meter/regulator with four alarm relays and large display, and SD8V panel meter/regulator with two alarm outputs from Termipol. By combining the measurement result display devices and the sensor, it is possible to continuously control the level with a single sensor. The sensor can be used in tanks for fuel, oil or high temperature liquids.
Example of application with pressure sensor and MG14 meter.
Set the input signal type Sn = 22 (4-20mA) → see instructions para. 3.
Set the minimum display value for the minimum value of the PUL input signal (we will set the display value for 0 mA, e.g. 0).
Set the maximum display value for the maximum value of the PUH input signal (we will set the display value for 20 mA e.g. 200).
Fig. Scaling of input signal for MG14 meter
Fig. Connection diagram with fuel level sensor PS
Example of application with pressure sensor and SD8V meter.
Set the input signal type 1nP = 4_20 (4-20 mA) → see instructions para. 1.
Set the minimum display value for the minimum value of the input signal FL = 0 (we will set the display value for 0 mA, e.g. 0).
Fig. Input signal scaling for SD8V meter
Set the maximum display value for the maximum input signal value FH = 3000 (we will set the display value for 20 mA e.g. 3000).
Fig. Wiring diagram with PS fuel level sensor